Sweet Cures® ’ own dedicated test Strips are a fast and easy-to-use indicator of 5 main elements (Leukocytes, Nitrites, Protein, pH, Blood) and help you know more precisely what is going on with your urine. You can manage the pH of your urine, check for nitrites and infection and monitor your progress.
Order before 5:30 pm for same day dispatch.
Professional Testing Strips
Monitor Waterfall D-Mannose progress
All Packaging Recyclable
UK Manufacturer – Sweet Cures
Quantity: 50 x Urine Test Strips
Questions? Call +44 01904-789559
While UTI Test 5V® is suitable for professional testing by medical professionals, it is also ideal for home use. We thus include a very easy to understand guide for people who are not at all technical, as well as the full medical information leaflet, for professionals who understand the technicalities.
Full instructions are inside the box.
The Test Strips are a fast and easy-to-use indicator of the infection condition of your urine. See your doctor or urologist for more detailed advice about interpreting the results of the test strips.
You can adjust your urine pH by making changes to your diet. See Making your urine more alkaline or try calling a dietician from the Yellow Pages.
The more colouration on the strip within the time frames indicated, the higher the possibility of bacteria being present, and at higher concentrations. To have your urine checked to find out detailed information about bacterial strains and kidney involvement, or for further interpretation of the results, see your doctor / medical expert.
While monitoring the health of your urine can help in understanding your progress, it may not represent the whole picture. UTI Test 5V® can tell you with a high degree of reliability if there is still infection present in your urine, but cannot tell you what species of bacteria are causing the symptoms. If natural remedies are unable to clear the infection, consider having your urine cultured by a competent clinic to identify the agents of infection.
*All customer reviews represent the personal opinion of the reviewer. Results may vary from person to person. All products are sold and have an intended use as Dietary Supplements. Nothing on this page is meant to constitute medical advice.
The test reveals the presence of granulocyte esterases. These esterases cleave an indoxyl ester, and the indoxyl so liberated reacts with a diazonium salt to produce a violet dye.
Leukocyte esterase results may be positive in the absence of observable cells if the leukocytes have lysed. Positive results may occasionally occur with random specimens from females due to contamination of the urine specimen by vaginal discharge. Elevated glucose concentrations (55-110 mmol/L) or high specific gravity may cause decreased test results. The presence of cephalexin, cephalothin, or tetracycline may cause decreased reactivity, and high levels of the drug may cause a false-negative reaction. The test area does not react with lymphocytes.
Based on the principle of Griess’s test and is specific to nitrites. Any degree of uniform pink colour development indicates positive results.
Positive results of the nitrites test suggest the presence of 105 (100,000) or more organisms per mL of urine, but colour development is not proportional to the number of bacteria present. A negative result does not in itself prove that there is no significant bacteriuria. Negative results may occur when urinary tract infections occur due to organisms that do not contain reductase to convert nitrate to nitrite. When urine has not been retained in the bladder long enough (4-8hrs) for reduction of nitrate to occur; or when dietary nitrate is absent, even if organisms containing reductase are present and bladder incubation is ample. Ascorbic acid concentrations of 2.8mmol/L or higher may cause false negative results with specimens containing nitrite ion concentrations of 35μmol/L or less.
Based on the principle of the protein error of a pH indicator.
The reagent area is sensitive to albumin. An elevated pH (up to 9) may affect the test. If the residue of disinfectant containing quaternary ammonium groups or chlorhexidine is present in the urine collection container, this may lead to a false positive result.
This test contains a mixed indicator which assures a marked change in colour between pH5 and pH8.5. It may be necessary to repeat this many times with a space of 3 hours between tests to determine the prevailing pH. High or low pH or wild swings in pH may indicate the presence of bacteriuria.
Haemoglobin and myoglobin catalyse the oxidation of the indicator by means of organic hydroperoxide contained in the test paper.
This test is highly sensitive to haemoglobin and thus complements microscopic examination. The sensitivity of a test diminishes in urine with high specific gravity. The test is equally sensitive to myoglobin as to haemoglobin (Haemoglobin concentration of 150-620 μg/L is approximately equivalent to 5-15 intact red occult blood cells per microlitre). Captopril and Lodine may also cause decreased reactivity. Blood is likely to found in the urine of menstruating females. Oxidizing contaminants, such as hypochlorite, may produce false positive results. Microbial peroxidase associated with urinary tract infection may cause a false positive reaction. Ascorbic acid concentrations of 2.0mmol/L or greater may cause false negatives at the trace levels.
Sensitivity is dependent upon the presence or absence of interfering specimens.
Based on the dry weight content of each reagent in 100 strips:
Leukocytes: indoxyl ester 1.4mg, diazonium salt 0.7mg;
Nitrite: sulfanilamide 0.65mg, N-(naphthyl)-ethylenediammonium dihydrochloride 0.45mg;
Protein: tetrabromphenol blue 0.36mg;
pH: methyl red 0.06mg, bromthymol blue 1.3mg;
Blood: cumene hydroperoxide 35.2mg, 3,3',5,5' -Tetramethylbenzidine 15.0mg.
1. HANDLING: Use a clean container to collect urine. False-positive readings for Blood can result from residues of strongly oxidising disinfectants in the specimen collection container. Do not add preservatives to the urine to be tested.
2. OPERATION: Avoid shaking the test strip in a specimen container to prevent incorrect results. Dip for approx. Two seconds. If the dipping time is too short or too long, the results may be unpredictable.
Collect urine in a clean container and test as soon as possible. Do not centrifuge. Avoid the use of urine preservatives. If testing cannot be performed within one hour after voiding, refrigerate the specimen immediately. Allow refrigerated specimen to return to room temperature before testing.
1. Remove one strip from the bottle at a time and replace cap tightly.
2. Immerse reagent areas of the strip in fresh, well-mixed urine. Remove the urine test strip immediately to avoid the dissolving out of reagent areas.
3. While removing, tap the strip against the rim of the urine container to remove excess urine. Blot the urine strip on a long edge, on absorbent paper. Avoid running over (contamination from adjacent reagent pads).
4. Compare each reagent area to its corresponding colour blocks, shown on the colour chart and read at times specified. Proper read-time is critical for optimal results.
5. Obtain results by direct colour chart comparison.
NOTE: Read reagent results between 1-2 minutes for differentiating a positive sample from a negative sample. Changes in colour after 2 minutes are of no diagnostic value.
For best results, confirm the performance of reagent strips by comparing with known negative and positive specimens or controls. This procedure is recommended after use of every 25 urine strips and for each change in urine strip containers and lot. User/labs should establish their own goals for performance standards.
Obtain results by direct comparison of the colour chart printed on the container label. The colour chart represents nominal values. Actual values will vary around the nominal values.
Improper storage may cause inadequate performance of test strips. Return to room temperature before use. Do not use deteriorated, discoloured or blackened test strips. Avoid contamination by volatile chemicals. Do not touch test papers of reagent strips.
On principle, Use multiple results for diagnosis or therapy, in the context of all other medical findings. Knowledge of the effects of drugs or their metabolites upon the individual tests is not yet complete. In doubtful cases, it is, therefore, advisable to repeat any testing after discontinuing a particular drug. Large amounts of ascorbic acid in the urine can produce artificially low to false-negative results when testing for blood and nitrites.
Store at temperatures between 2 C to 30 C avoiding humidity, direct sunlight, or heat. Store only in the original bottle. Do not remove desiccants. Do not remove a strip from the bottle until immediately before its use. Replace cap quickly and tightly after removing reagent strip. After opening use any remaining test strips within three months. Do not use reagent strips after the expiry date printed on the label.
Sweet Cures ’ Anna talks about her battle with cystitis. Read More
New Waterfall D-Mannose Lemon is a delicious and refreshing approach to natural bladder health.
Probiotic Blend supports the gut and can be an aid to wellbeing when consumed as part of a healthy diet.
Waterfall D-Mannose supports a Healthy Bladder.
A food supplement which provides support when trying to lose weight.
A powerful anti-microbial with 80% Carvacrol and 20% organic olive oil which has properties to support natural defences in the fight against bacteria, fungi, parasites and viruses.
A food supplement which supports healthy energy levels.